Platform comparison of binding kinetics of Nivolumab
The pharmacological response of a biological drug is profoundly influenced by its non-specific binding to serum proteins. Interactions with blood proteins affect conformational changes and modulate the efficacy of the drug, which in some extreme situations cause antagonistic drugs to lose their capacity to bind to their receptors. How strongly a drug protein binds to its target molecule alters the drug’s potency, dosing regimen and degree of on-target side effects. Unfortunately, the platforms typically used to measure binding strength, such as those relying on Surface Plasmon resonance (SPR) or Bio-Layer Interferometry (BLI), are unable to consistently and reliably measure protein binding kinetics in a physiologically relevant system.
In contrast, the MagArray Protein Binding Kinetics platform can measure protein binding kinetics precisely, reproducibly, and with great sensitivity in physiologically relevant matrices. A recent comparison study of the kinetics of Nivolumab binding with PD-1 in increasing concentrations of serum, demonstrated the MagArray platform can measure binding on rates and off rates when the SPR and BLI systems cannot.
The MagArray platform was able to provide a reliable and rapid approach to obtaining binding kinetics parameters in physiologically relevant matrices while the SPR platform could not measure the kinetics in a matrix containing more than 0.5% serum. The BLI platform could not measure binding kinetics in the presence of any amount of serum.
Given the profound impact that cancer immunotherapies are having, and the rapid increase in the number of antibody checkpoint inhibitors being investigated, the importance of examining their binding kinetics in a physiologically relevant matrix is critical to understanding exactly how such immuno-oncology inhibitors are likely to behave in-vivo.
To learn more about how the MagArray Protein Kinetics platform can help your drug discovery and characterization efforts, click here, or reach out to Kalidip Choudhury, Ph.D
MagArray Lifesciences Kalidip.firstname.lastname@example.org
Giant Magnetic Resonance (GMR) Assay Principles
Our platform utilizes a sandwich assay in which the target antigen is sandwiched between two antigens, one bound to the magnetic sensor and the other to a superparamagnetic nano particle. Under an external magnetic field, the nanoparticles magnetize and their presence or absence can be detected by the sensor. Using chips with 64 GMR sensors we show rapid multiplex protein detection with a liner dynamic range of over 5-6 orders of magnitude for a diverse range of biological fluids in real time.
*Based on standard LBA process with magnetic nanoparticles used as labels
Instrument and chip
GMR Multiplex Chip
- 80 individual sensors on each chip
- 8 chips with 640 sensors total
- Each sensor can be individually spotted
- 80 biomarkers with controls for the multiplex chip
GMR Reader Unit
- Rugged and reliable
- No optics meaning no calibration of lasers, mirrors etc
- No fluidics means no pumps, valves, etc
- Fully automated
- Hands free incubation
- Bar code system to minimize errors
- No wash necessary
- Flexible assay protocols
- Run time of 1 hour for most sensitive assay
- Run time of 5-10 minutes for express assays
MagArray has 5-6 orders of magnitude greater dynamic range and a 1000 times more sensitive than ELISA.
Compared to ELISA, the MagArray platform is capable of detecting CEA at significantly lower concentrations and at 5-6 orders of magnitude greater dynamic range utilizing the same Ab pair
Matrix insensitive detection in different buffers.
A panel of eight human tumor markers with BSA negative control and epoxy control indicates matrix insensitive detection when shifting from PBS to mouse serum to lysis buffer
Ligand Binding Assay
Our GMR platform is more sensitive than SPR and insensitive to buffer conditions
MagArray platform at least 1000 times more sensitive than Biacore/SPR
MagArray platform insensitive to pH and salinity*
The effect of matrix on protein binding between SPR and MagArray
Buffer: 0.01M HEPES, pH 7.4, 0.15M NaCl, 3mM EDTA
Regeneration: 10mM Gly-HCL, pH 2.5
Analyte: TSH, Ligand: Anti-TSH ab
Biacore X 100
- The same binding pair and buffer conditions were used to compare between MagArray and Biacore
- Plasma was used as a matrix and diluted by PBS to 80%, 50%, 10%, 1%, 0.1%
- MagArray platform kinetic measurements were not affected by matrix
- Biacore did not provide reliable results above 0.1% matrix
- Truly a transformational platform that helps to bridge the gap between in-vitro and in-vivo.
- Matrix insensitive technology for measuring biomolecules in blood, plasma, serum, saliva and urine.
- 1000 times more sensitive than leading SPR technologies in measuring protein-protein interactions.
- Partnership with Hitachi Hi-Technologies allows expansion of platform for unmet needs in biological assays.
- R. S. Gaster, D. A. Hall, C. H. Nielsen, S. J. Osterfeld, H. Yu, K. E. Mach, R. J. Wilson, B. Murmann, J. C. Liao, S. S. Gambhir & S. X. Wang, “Matrix-insensitive protein assays push the limits of biosensors in medicine,” Nature Medicine, Oct. 11 (online publication), 2009.
Vermesh O, Aalipour A, T. Ge J, Saenz Y, Guo Y, Alam IS, Park S-min, Adelson CN, Mitsutake Y, Vilches-Moure J et al.. 2018. An intravascular magnetic wire for the high-throughput retrieval of circulating tumour cells in vivo. Nature Biomedical Engineering. 2:696-705.Google Scholar
Lee J.R., Appelmann I, Miething C, Shultz T.O., Ruderman D, Kim D, Mallick P, Lowe S.W., Wang S.X.. 2018. Longitudinal Multiplexed Measurement of Quantitative Proteomic Signatures in Mouse Lymphoma Models Using Magneto-Nanosensors. Theranostics. 8(5)Google Scholar
Ng E, Yao C, Shultz TO, Ross-Howe S, Wang SX. 2018. Magneto-nanosensor Smartphone Platform for the Detection of HIV and Leukocytosis at Point-of-Care. Nanomedicine: Nanotechnology, Biology and Medicine. Google Scholar
Beggs M., Yu H., Carbonell L., Wang S.X., Vachani A.. 2018. Validation of Plasma TIMP-1 to Identify Lung Cancer in Smokers. D99. CLINICALLY INFORMATIVE BIOMARKERS IN LUNG CANCER: A NEEDLE IN A HAYSTACK. :A7415-A7415.Google Scholar
Lee J-R, Chan CT, Ruderman D, Chuang H-Y, Gaster RS, Atallah M, Mallick P, Lowe SW, Gambhir SS, Wang SX. 2017. Longitudinal Monitoring of Antibody Responses against Tumor Cells Using Magneto-nanosensors with a Nanoliter of Blood. Nano Letters. 17:6644-6652.Google Scholar
Rizzi G, Lee J-R, Dahl C, Guldberg P, Dufva M, Wang SX, Hansen MF. 2017. Simultaneous Profiling of DNA Mutation and Methylation by Melting Analysis Using Magnetoresistive Biosensor Array. ACS Nano. 11:8864-8870.Google Scholar
Ng E, Nadeau KC, Wang SX. 2016. Giant magnetoresistive sensor array for sensitive and specific multiplexed food allergen detection.. Biosens Bioelectron. 80:359-365.PubMed
Lee J-R, D Haddon J, Gupta N, Price JV, Credo GM, Diep VK, Kim K, Hall DA, Baechler EC, Petri M et al.. 2016. High-Resolution Analysis of Antibodies to Post-Translational Modifications Using Peptide Nanosensor Microarrays.. ACS Nano. PubMed
Brett E, Zielins ER, Luan A, Ooi CChun, Shailendra S, Atashroo D, Menon S, Blackshear C, Flacco J, Quarto N et al.. 2016. Magnetic Nanoparticle‐Based Upregulation of B‐Cell Lymphoma 2 Enhances Bone Regeneration. STEM CELLS Translational Medicine. 6:151-160.Google Scholar
Lee J-R, Bechstein DJB, Ooi CChun, Patel A, Gaster RS, Ng E, Gonzalez LC, Wang SX. 2016. Magneto-nanosensor platform for probing low-affinity protein-protein interactions and identification of a low-affinity PD-L1/PD-L2 interaction.. Nat Commun. 7:12220.PubMed
Park S-min, Wong DJ, Ooi CChun, Kurtz DM, Vermesh O, Aalipour A, Suh S, Pian KL, Chabon JJ, Lee SHun et al.. 2016. Molecular profiling of single circulating tumor cells from lung cancer patients. Proceedings of the National Academy of Sciences. 113:E8379-E8386.Google Scholar
Lee J-R, D Haddon J, Wand HE, Price JV, Diep VK, Hall DA, Petri M, Baechler EC, Balboni IM, Utz PJ et al.. 2016. Multiplex giant magnetoresistive biosensor microarrays identify interferon-associated autoantibodies in systemic lupus erythematosus.. Sci Rep. 6:27623.PubMed
Earhart CM, Hughes CE, Gaster RS, Ooi CChun, Wilson RJ, Zhou LY, Humke EW, Xu L, Wong DJ, Willingham SB et al.. 2014. Isolation and mutational analysis of circulating tumor cells from lung cancer patients with magnetic sifters and biochips.. Lab Chip. 14(1):78-88.PubMed